ESCORT 3136A PDF
View and Download Escort A operation manual online. Count Multimeter. A Multimeter pdf manual download. Here you will find a large selection of new and used Escort A listed at special prices. Although some items match several categories, all of them are listed. Section 7. GPIB Remote Operation. Introduction. Description of the GPIB. GPIB Interface Parameters Setup. Commands Summary. 72 .
|Country:||Papua New Guinea|
|Published (Last):||18 January 2007|
|PDF File Size:||14.3 Mb|
|ePub File Size:||11.34 Mb|
|Price:||Free* [*Free Regsitration Required]|
Escort 3136A Operation Manual
Introduction Section 4 Measurement Application Examples Introduction Section 4 describes some advanced features and applications that help the user to operate the meter more effectively. Calibration Mode Button 9. If another positive range is specified, the nearest compatible range is set. Comments to this Manuals Your Name. Resets the multimeter’s parameters to their default values.
Since it is the default, 313a6 command can be shortened in: After this command, the secondary display still indicates signal source as the frequency measurement is on primary display. The parameters set on the device must be these ones default. Auto range AUTO 9.
ESCORT A OPERATION MANUAL Pdf Download.
Query Commands Query Commands R0 command R0 command is used for requesting the meter to return its current status. Operating in Explosive Atmospheres The meter does not provide explosion protection for explosive gasses or arcing components. Arithmetic Function Selection 5. Page 10 To avoid explosion, do not operate this product in an explosive atmosphere. Dual Display Operation Examples Dual Display Operation Examples This section will describe some practical operations of using dual display features.
FREQuency [  escorrt, 2 ] Description: Connect a resistor under test to and COM input terminals as shown below: Some display assignments may switch the actual displays in order to comply esccort possible pairs.
The escorr mode will rotate as the following sequences if keep pressing the key: The key “local” is also disabled. Input Terminals Input Terminals The input terminals, shown in Figure are located on the left side of the front panel.
Select the proper function for your measurement. As configuration data for IEEE address, RS baud rate, data bit, stop bit, parity, echo and so on are stored in the non-volatile memory, they are not changed when power is cycled off and on until the configurations are changed by escogt user.
Relative mode REL Selects the source of the start trigger signal.
Download file %% from thread %Multimetr Escort – schemat potrzebny!%
Grounding the Meter The meter is grounded through power cord. To display AC current in primary display and DC current in secondary display or vise versus while testing a rectifier circuit, user may check the DC current component and its AC ripples by taking a single meter.
To print the manual completely, please, download it. Page 87 Table This function will be used for lock range. Sets the brightness of the displays. Strings are case unsensitive so for instance “5m” could either be used to specify the 5mV range, or instead of “5M” for the 5 Mohm range, depending on the context. Key Commands There are 16 pushbutton keys on the front panel of the meter.
Measurement Function Selection 7. Instructions Of Command Sets If RS of the meter is under print-only mode, the meter will print out the measured data when the measurement cycle is completed. Measuring AC and DC current on a rectification circuit. BUS – Selects a bus command and enter single trigger mode.
Don’t have an account? Specifying a numerical value is also possible.
Alternative equipment may be used as long as the accuracy is at least as good as those listed. Any command except “local” sent to the device through the serial port automatically turns its adressing mode to “REMOTE”, if it was formerly in “local” mode mmA lock: First scan the Tables of contents to be familiar with the outline of the manual. Turns “on” the trigger mode. Place the meter to power-on-reset state, but no affect 3136aa items: To avoid electric shock or injury, grounding wire in the power line cord must be connected.
OPC – Operation complete. Options And Accessories Options and Accessories At the moment, one option is available for the meter, which option can be installed at the factory and a field installable retrofit kit option is also available: DBM – dBm calculation.
Page 65 Table Section 3 Front Panel Operation Introduction This section provides a 3136 description of each operation that can be performed by using the pushbuttons on the front panel.
ESCORT-3136A = FI-3136A multimeter
Turns “off” the trigger mode. Page 67 R1 command After executing R1 command, the escortt will return the existing readings of primary display. This display shows measuring readings, annunciator, and messages. Selecting Local Operation Mode Selecting Local Operation Mode to return the operation control priority from remote mode computer controlled to local mode user controlled.