COBAS C111 ROCHE PDF
Consult Roche’s entire cobas c analyzer catalogue on MedicalExpo. Page: 1 / The cobas c , a new low volume bench-top analyzer from Roche Diagnostics (Indianapolis, IN), was evaluated to determine whether the. Roche Cobas c – Chemistry Analyzer Information.
|Published (Last):||2 June 2006|
|PDF File Size:||15.90 Mb|
|ePub File Size:||7.87 Mb|
|Price:||Free* [*Free Regsitration Required]|
The results of this study support the use of the cobas c in the small laboratory niche environment of less than 60 patients per day. rlche
Inquire Here Sample Cycle Time: Analyte-specific sample sets were picked to challenge the measuring range. Citing articles via Web of Science The intuitive user interface, barcode reader for samples, reagents and applications, and host connectivity option make the system very attractive in situations where operators may have minimal supervision and moderate skill sets.
If imprecision could be reduced for DBIL, it is very possible that it could reach world-class performance. Inquire Here Sample Input Type: Email alerts New issue alert. You can add Community Subscriptions in the search bar that says “Subscribe to more communities Inquire Here Reagent Type: A serum pool was aliquoted into 5 cups for each system and 5 values obtained for each analyte from which a mean and standard deviation SD were derived.
Cobas c – Roche – Chemistry Analyzer
Overall, the average sigma for 21 methods excluding ones that provided infinity sigma results due to 0 imprecision is Inquire Here Reagent Capacity: The DBIL slope of 0.
The calculations were performed using James O. Como procedo para solucionar? Only 1 method DBIL demonstrates some challenges sigma 2. Although the cobas c can complete the testing of panels like the 14 assay Comprehensive Metabolic Panel, the throughput of the instrument indicates workflow for such testing should be carefully managed to allow instrument time to serve more patients who need stat testing of much fewer analytes.
Assays Total assays tested: Additionally the performance of the cobas c was evaluated using sigma metrics.
Roche Cobas c111
Final Rule for Current Labs Published. Seventeen out of 23 tested methods performed rovhe the world-class level, 2 demonstrated excellent performance, and 3 demonstrated good performance.
It furthers the University’s objective of excellence in research, scholarship, and education by publishing worldwide. The cobas c is a random access analyzer with stat capability offering a mix of enzymes, substrates, serum proteins, and electrolytes suitable for a small hospital or physician office laboratory.
Also included is a base matrix 0. Improving Urine Sediment Analysis. Additionally correlation studies, using patient serum samples, were used to determine inter-instrument comparability between 2 cobas c instruments and accuracy relative to the standardized assays at LabCorp.
Related articles in Web of Science Google Scholar. Standby to operation takes approximately 3 minutes. Oxford University Press is a department of the University of Oxford.
Roche Cobas C – Chemistry Analyzer – Diamond Diagnostics
Inquire Here Water Consumption: Interference from bilirubin, lipids, or hemolysis was not evaluated as part of this study. Inquire Here Standby Mode: Inquire C11 Reagent Delivery: Inquire Here Test Method: Inquire Here for a complete list! The objective is to provide consistent patient results with the same reference ranges throughout the network regardless of the point of service.
Table 5 shows the cobas c is very close to complete world-class performance. This allows standardization of results across the LabCorp network or any other vertically integrated health care system that also selects the larger Roche platforms c11 their core laboratories.
Precision, linearity, method comparison to another large Roche system, inter-instrument comparability, potential carryover, and time to result studies met or exceeded expectations.
Inquire Here Water Requirement: Cobas C Brochure. c1111
Roche cobas c 111
Daily start-up and system checks are prompted by the user interface and involve: The cobas c analyzer is a tabletop system designed to set the next gold standard in automated clinical chemistry testing for small laboratories. Forums Documents Parts Videos News.
Inquire Here On-Board Refrigeration: You can adjust your Community Subscriptions in Settings. Additionally, the cobas c uses the same bulk reagents as larger Roche systems currently used by LabCorp.
Inter-instrument recovery passed for all analytes data not shown. Initial inspection of the precision data shows some between run coefficients of variation CVs that are unexpectedly less than the within run CV.
D-dimer and homocysteine are planned for later release. They are available as a liquid; 5—6 bottle sets from Maine Standards. Serum, Plasma, Urine Maximum Throughput: Cogas system uses the same calibrators, controls, and reagents available on the larger Roche platforms and performance met or c1111 comparative expectations.
Inquire Here Optical Source: Inter-assay precision was performed by running each control level twice per day over 5 days and included 2 to 3 calibrations per analyte.